The control panel is equipped with a microprocessor-controlled digital timer, to manually select the exposure time. Also, the automatic selection allows the operator from choosing among 15 pre-set times, according to the patient type (Large adult, Medium adult, and child) and the zone to be exposed. It also admits the selection of the radiography type (digital or conventional). The preprogrammed times can be adjusted by the user according to the film type and development process.
Manufacturer FIAD INTERNATIONAL S.A.S.
Current input 8.2 A
Power consumption 0.8 kVA @120V
Voltage input regulation 3%
Circuit brakes 20 A
Fuses 20 A – 250 V
Line Voltage 110 V – 220 V
Line frequency 50 Hz – 60 Hz
Timer (Selected exposure times) 0.1 to 1.3 seconds Timer (Corrected exposure times) 0.1 to 1.3 seconds Control Panel dimensions 9 ¼ H x 7´´ W x 3´´D Wall plate dimensions 12´ H x 9´ W x 1´ D
TUBE HEAD SPECIFICATIONS
Voltage output 60-70 kVp +- 7% High voltage circuit type 108-132 VAC, single-phase, self-rectified Current output 8 mA +- 1 mA Power 0.8 kW Total filtration 2,5 mm aluminum equivalent to 70 kVp Time interval between exposure (duty cycle) 32 times the exposure time / 1:212 X ray limiting device (Collimator) Incorporated to the tube head assembly (non-removable) Minimum focus-skin distance 20 cm (8´´) Maximun symmetric radiation field @20cm 2 3/8´´ (6cm) Radiation absorbed at 1m <50 mr/h, duty cycle 1:32 Technical factors of the absorbed 70 kVp, 8 mA, 1.3 second.